NOTE: IMPORTANT NOTE: DO NOT PLUG ANY EQUIPMENT INTO A WALL CHARGER THAT ACCEPTS A USB CABLE. ALL WARRANTIES WILL BE VOIDED IF UNAUTHORIZED CHARGING EQUIPMENT IS USED WITH INSiGHT TECHNOLOGIES.
INSiGHT Scanning Technologies will arrive fully charged and ready for use. All instruments in the wireless INSiGHT Scanning Technology suite have internal batteries that have an average minimum battery life of 90 minutes of continuous use.
Battery level is viewable at the top of each respective instrument’s scan screen.
NOTE: You should not let your equipment fully lose charge. Charge your equipment when the battery level falls below 25%.
Each instrument except spineSENSE includes a cable that has a charging jack on one end and a USB connector on the other. Plug the charging jack into the instrument and the USB into a computer that is turned on to charge.
Preferred charging option: The INSiGHT Bridge offers two USB ports. Two instruments can be charged simultaneously using this method by using the authorized charging wires and plugging them from the instrument into the USB port Note: If you choose to charge all instruments at the same time, you will need added USB ports and should charge them using a separate laptop or computer with USB ports.
A powered USB hub may be used UNDER STRICT CONDITIONS.Please consider these points:
- Use only the charging cables, provided to you by CLA, when you purchased your wireless technologies (other charging cables or power supplies with similar connectors should never be used)
- If a USB port on a PC/laptop is not available for charging and a USB wall charger or powered USB hub is to be used, it is important that a reputable brand named device (eg. Apple, Anchor) is selected
- Laptops contain better voltage monitoring and provide a safer charging experience for CLA devices devices with an “Express” charging feature that provide a faster charge for devices such as cell phones and laptops should NOT be used for CLA devices
|Equipment||Equipment Name||Jack Location||Charging Status|
||neuroTHERMAL||Bottom of handle||Yellow LED light when charging. Green LED light when fully charged.|
||neuroCORE||Bottom of handle||“Charger Attached” on the unit’s screen.|
|neuroPULSE||Front of unit||Flashing green LED when charging, Solid green LED light when charged.|
||spineROM||Base Unit (Unplug from Satellite to charge)||Yellow LED light when charging. Green LED light when fully charged.|
||spineSENSE||N/A||The spineSENSE plugs into the neuroPULSE, and therefore, the neuroPULSE must be charged.|